What is the concentration of chloride ions in the test solution and in the fall-out?
EasyPlus Chloride Titrator
Titration instead of refractometric, the chemical alternative for frequent measuring.
Depending on the adjustment of the salt spray cabinet the chloride concentration of the fall-out can deviate from the original content in the test solution tank. For that reason we recommend to determin the chloride concentration also in the collected fall-out, particularly when altering the temperature of the humidifier. The EasyPlus chloride titrator is a practical alternative to the frequently used refractometers (it is also offered by VLM).